|1.||Chen, Tao: 1 article (01/2012)|
|2.||Lu, Tingli: 1 article (01/2012)|
|3.||Ma, Yufan: 1 article (01/2012)|
|4.||Wang, Zhao: 1 article (01/2012)|
|5.||Zhang, Yang: 1 article (01/2012)|
|6.||Zhong, Pei: 1 article (11/2004)|
|7.||Cocks, Franklin H: 1 article (11/2004)|
|8.||Preminger, Glenn M: 1 article (11/2004)|
|9.||Zhou, Yufeng: 1 article (11/2004)|
|10.||Zhong, P: 1 article (03/2000)|
11/01/2004 - "Second, a piezoelectric annular array (PEAA) generator (f = 230 kHz and F = 150 mm) was used to produce an auxiliary shock wave of approximately 13 MPa in peak pressure (at 4 kV output voltage) to intensify the collapse of LSW induced bubbles near the target stone for improved comminution efficiency. "
03/01/2000 - "A maximum increment of 60% to 80% in stone fragmentation was achieved when the PEAA-generated shock wave arrives near the collapse of the bubbles. "
03/01/2000 - "Hence, the PEAA-generated shock wave could be used to interact with cavitation bubbles induced by the EH source at various stages of their oscillation. "
03/01/2000 - "At a discharge voltage of 15 kV, the PEAA generator produces a shock wave with a peak positive pressure of 8.2 MPa, a positive half cycle duration of 2.9 micros, and a -6-dB beam width of 5 mm. The trigger of the PEAA generator was controlled via fiberoptic link with reference to the spark discharge of the EH generator. "
03/01/2000 - "To control the collapse of cavitation bubbles induced during shock-wave lithotripsy (SWL), a piezoelectric annular array (PEAA) shock-wave generator was fabricated and combined with an experimental electrohydraulic (EH) shock-wave lithotripter with a truncated HM-3 reflector. "
11/01/2004 - "= 0.2 mm) was produced after 30 shocks by the original HM-3 reflector at 20 kV. In comparison no vessel rupture could be produced after 200 shocks using the upgraded reflector at 22 kV or the PEAA generator at 4 kV. Using cylindrical BegoStone phantoms (Bego USA, Smithfield, Rhode Island) stone comminution efficiencies (mean +/- sd) after 1,500 shocks produced by the original and upgraded HM-3 reflectors, and the combined PEAA/upgraded HM-3 system, were 81.3% +/- 3.5%, 90.1% +/- 4.3% and 95.2% +/- 3.3%, respectively (p<0.05). "
|2.||glucuronyl glucosamine glycan sulfate (Vessel)
|1.||Lithotripsy (Extracorporeal Shockwave Lithotripsy)